All information products included in https://us-cert.cisa.gov/ics are provided "as is" for informational purposes only. The Department of Homeland Security (DHS) does not provide any warranties of any kind regarding any information contained within. DHS does not endorse any commercial product or service, referenced in this product or otherwise. Further dissemination of this product is governed by the Traffic Light Protocol (TLP) marking in the header. For more information about TLP, see https://us-cert.cisa.gov/tlp/.
1. EXECUTIVE SUMMARY
- CVSS v3 5.3
- Vendor: Philips
- Equipment: Veradius Unity, Pulsera, and Endura Dual WAN Router
- Vulnerability: Inadequate Encryption Strength
2. RISK EVALUATION
Successful exploitation of this vulnerability could compromise the management interface of the front end router impacting the availability of data transfer via wireless communication.
3. TECHNICAL DETAILS
3.1 AFFECTED PRODUCTS
Philips reports the following Veradius Unity, Pulsera, and Endura Dual WAN Routers are affected:
- Veradius Unity (718132) with wireless option (shipped between 2016-August 2018)
- Veradius Unity (718132) with ViewForum option (shipped between 2016-August 2018)
- Pulsera (718095) and Endura (718075) with wireless option (shipped between 26-June-2017 through 07-August 2018)
- Pulsera (718095) and Endura (718075) with ViewForum option (shipped between 26-June-2017 through 07-August 2018)
3.2 VULNERABILITY OVERVIEW
The router software uses an encryption scheme that is not strong enough for the level of protection required.
- CRITICAL INFRASTRUCTURE SECTORS: Healthcare and Public Health
- COUNTRIES/AREAS DEPLOYED: Worldwide
- COMPANY HEADQUARTERS LOCATION: Netherlands
Daniel Yagudayev from New York Presbyterian Hospital reported this vulnerability to Philips.
Philips has a solution available for users who have the wireless or ViewForum option in their product to update the configuration of the Dual WAN router.
Philips contact information is available at the following link: Philips Customer Service.
Please see the Philips product security website for the latest security information for Philips products.
CISA recommends users take defensive measures to minimize the risk of exploitation of this vulnerability. Specifically, users should:
- Restrict system access to authorized personnel only and follow a least privilege approach.
- Apply defense-in-depth strategies.
- Disable unnecessary accounts and services.
- Where additional information is needed, refer to existing cybersecurity in medical device guidance issued by the FDA.
CISA reminds organizations to perform proper impact analysis and risk assessment prior to deploying defensive measures.
CISA also provides a section for control systems security recommended practices on the ICS webpage on us-cert.gov. Several recommended practices are available for reading and download, including Improving Industrial Control Systems Cybersecurity with Defense-in-Depth Strategies.
Additional mitigation guidance and recommended practices are publicly available on the ICS webpage on us-cert.gov in the Technical Information Paper, ICS-TIP-12-146-01B--Targeted Cyber Intrusion Detection and Mitigation Strategies.
Organizations observing any suspected malicious activity should follow their established internal procedures and report their findings to CISA for tracking and correlation against other incidents.
No known public exploits specifically target this vulnerability. This vulnerability is not exploitable remotely. High skill level is needed to exploit.
For any questions related to this report, please contact the CISA at:
Toll Free: 1-888-282-0870
CISA continuously strives to improve its products and services. You can help by choosing one of the links below to provide feedback about this product.
Please share your thoughts.
We recently updated our anonymous product survey; we'd welcome your feedback.