All information products included in https://us-cert.gov/ics are provided "as is" for informational purposes only. The Department of Homeland Security (DHS) does not provide any warranties of any kind regarding any information contained within. DHS does not endorse any commercial product or service, referenced in this product or otherwise. Further dissemination of this product is governed by the Traffic Light Protocol (TLP) marking in the header. For more information about TLP, see https://www.us-cert.gov/tlp/.
1. EXECUTIVE SUMMARY
- CVSS v3 9.8
- ATTENTION: Exploitable remotely/low skill level to exploit
- Vendor: Medtronic
- Equipment: Valleylab FT10, Valleylab FX8
- Vulnerabilities: Use of Hard-coded Credentials, Reversible One-way Hash, Improper Input Validation
2. RISK EVALUATION
Successful exploitation of these vulnerabilities may allow an attacker to overwrite files or remotely execute code, resulting in a remote, non-root shell on the affected products. By default, the network connections on these devices are disabled. Additionally, the Ethernet port is disabled upon reboot. However, it is known that network connectivity is often enabled.
3. TECHNICAL DETAILS
3.1 AFFECTED PRODUCTS
The following Medtronic Valleylab products, are affected:
- Valleylab Exchange Client, Version 3.4 and below
- Valleylab FT10 Energy Platform (VLFT10GEN) software Version 4.0.0 and below
- Valleylab FX8 Energy Platform (VLFX8GEN) software Version 1.1.0 and below
3.2 VULNERABILITY OVERVIEW
The affected products use multiple sets of hard-coded credentials. If discovered, they can be used to read files on the device.
The affected products use the descrypt algorithm for OS password hashing. While interactive, network-based logons are disabled, and attackers can use the other vulnerabilities within this report to obtain local shell access and access these hashes.
The affected products use a vulnerable version of the rssh utility in order to facilitate file uploads. This could provide an attacker with administrative access to files or the ability to execute arbitrary code.
- CRITICAL INFRASTRUCTURE SECTORS: Healthcare and Public Health
- COUNTRIES/AREAS DEPLOYED: Worldwide
- COMPANY HEADQUARTERS LOCATION: Ireland
Medtronic reported these vulnerabilities to CISA.
Software patches are currently available for the FT10 platform and will be available in early 2020 for the FX8 platform. Until these updates can be applied, Medtronic recommends to either disconnect affected products from IP networks or to segregate those networks, such that the devices are not accessible from an untrusted network (e.g., Internet). Patches can be downloaded at the following location:
Medtronic has released additional patient focused information, at the following location:
CISA recommends users take defensive measures to minimize the risk of exploitation of these vulnerabilities. Specifically, users should:
- Minimize network exposure for all medical devices and/or systems.
- Locate medical devices behind firewalls and isolate them where possible.
- Restrict system access to authorized personnel only and follow a least privilege approach.
- Apply defense-in-depth strategies.
- Disable any unnecessary accounts, protocols and services.
- Where additional information is needed, refer to existing cybersecurity in medical device guidance issued by the FDA at the following location: https://www.fda.gov/medical-devices/digital-health/cybersecurity
CISA reminds organizations to perform proper impact analysis and risk assessment prior to deploying defensive measures.
CISA also provides a section for control systems security recommended practices on the ICS webpage on us-cert.gov. Several recommended practices are available for reading and download, including Improving Industrial Control Systems Cybersecurity with Defense-in-Depth Strategies.
Additional mitigation guidance and recommended practices are publicly available on the ICS webpage on us-cert.gov in the Technical Information Paper, ICS-TIP-12-146-01B--Targeted Cyber Intrusion Detection and Mitigation Strategies.
Organizations observing any suspected malicious activity should follow their established internal procedures and report their findings to CISA for tracking and correlation against other incidents.
No known public exploits specifically target these vulnerabilities.
For any questions related to this report, please contact the CISA at:
Toll Free: 1-888-282-0870
CISA continuously strives to improve its products and services. You can help by choosing one of the links below to provide feedback about this product.