All information products included in https://us-cert.cisa.gov/ics are provided "as is" for informational purposes only. The Department of Homeland Security (DHS) does not provide any warranties of any kind regarding any information contained within. DHS does not endorse any commercial product or service, referenced in this product or otherwise. Further dissemination of this product is governed by the Traffic Light Protocol (TLP) marking in the header. For more information about TLP, see https://us-cert.cisa.gov/tlp/.
1. EXECUTIVE SUMMARY
- CVSS v3 5.3
- ATTENTION: Exploitable remotely/low skill level to exploit/public exploits are available
- Vendor: Philips
- Equipment: Philips Alice 6 System product
- Vulnerabilities: Improper Authentication, Missing Encryption of Sensitive Data
2. UPDATE INFORMATION
This updated advisory is a follow-up to the updated advisory titled ICSMA-18-086-01 Philips Alice 6 Vulnerabilities (Update A) that was published December 11, 2018, on the NCCIC/ICS-CERT website.
3. RISK EVALUATION
Successful exploitation may allow an attacker to gain visibility to usernames/passwords and personal data. Insufficient encryption and cryptographic integrity checks can lead to altered, corrupted, or disclosed sensitive data. Disclosure of personal data can occur by replacing a trusted node with a malicious node.
4. TECHNICAL DETAILS
4.1 AFFECTED PRODUCTS
The following versions of the Philips Alice 6 System are affected:
- Version R8.0.3 or prior
4.2 VULNERABILITY OVERVIEW
When an actor claims to have a given identity, the software does not prove or insufficiently proves that the claim is correct. This weakness can lead to the exposure of resources or functionality to unintended actors, possibly providing attackers with sensitive information or the ability to execute arbitrary code.
The lack of proper data encryption passes up the guarantees of confidentiality, integrity, and accountability that properly implemented encryption conveys.
- CRITICAL INFRASTRUCTURE SECTORS: Healthcare and Public Health
- COUNTRIES/AREAS DEPLOYED: Worldwide
- COMPANY HEADQUARTERS LOCATION: Netherlands
Philips reported these vulnerabilities to NCCIC.
Philips will notify users of the identified vulnerabilities and will coordinate with users to schedule updates. Philips is scheduled to release a new product version and supporting product documentation for Q1 of 2019. For all users of the Alice 6 System product, Version R8.0.3 or prior, Philips will update the devices to R8.0.4. Philips encourages users to use Philips validated and authorized changes only for the Alice 6 device supported by Philips’ authorized personnel, or under Philips’ explicit published directions for patches, updates, or releases.
As an interim mitigation to the vulnerabilities until the update can be applied, Philips recommends that users:
- Ensure that network security best practices are implemented, and
- Limit network access to Alice 6 in accordance with product documentation.
Users with questions regarding their specific Alice 6 installations should contact their local Philips service support team or their regional Alice 6 service support. Contact information is available at the following location:
NCCIC recommends users take defensive measures to minimize the risk of exploitation of this vulnerability. Specifically, users should:
- Minimize network exposure for all control system devices and/or systems, and ensure that they are not accessible from the Internet.
- Locate control system networks and remote devices behind firewalls, and isolate them from the business network.
- When remote access is required, use secure methods, such as Virtual Private Networks (VPNs), recognizing that VPNs may have vulnerabilities and should be updated to the most current version available. Also recognize that VPN is only as secure as the connected devices.
NCCIC reminds organizations to perform proper impact analysis and risk assessment prior to deploying defensive measures.
NCCIC also provides a section for control systems security recommended practices on the ICS-CERT web page. Several recommended practices are available for reading and download, including Improving Industrial Control Systems Cybersecurity with Defense-in-Depth Strategies.
Additional mitigation guidance and recommended practices are publicly available on the ICS-CERT website in the Technical Information Paper, ICS-TIP-12-146-01B--Targeted Cyber Intrusion Detection and Mitigation Strategies.
Organizations observing any suspected malicious activity should follow their established internal procedures and report their findings to NCCIC for tracking and correlation against other incidents.
For any questions related to this report, please contact the CISA at:
Toll Free: 1-888-282-0870
CISA continuously strives to improve its products and services. You can help by choosing one of the links below to provide feedback about this product.
Please share your thoughts.
We recently updated our anonymous product survey; we'd welcome your feedback.